MaxBite Wireless Intraoral Lighting System
* Designed for optimal patient comfort
* “Bite block” keeps the patients’ mouths open
* “Bite blocks” are autoclavable
* Suction function (optional)
• MODEL : portable MaxBite / (Intra oral) Dental light
• INPUT : AC 100~240 VAC / OUTPUT : DC 5V / 1A
• Battery charging time : 1 Hours / consecutive use after battery charging 1 Hours
• LED luminance over 5,000 lux / Storage Temperature -20℃ ~ 70℃
• Operating Temperature -10℃ ~ 40℃ / Humidity 10% ~ 80%
• Product Classification : Dental light
• Pack size : Portable MaxBite(box) (L)285 x (W)150 x (H)75 /(mm)
• Delivery lead time : 4weeks / 1,000box
▣ Medical examination of teeth and consultation of patient.
▣ Dental examination before and after implant, scaling and tooth whitening and reformation
▣ Dental examination by travel or by home visit, and external dental treatment for the old or disabled person
▣ Suction (optional)
1) Select Biteblock patient will fit in the mouth.
2) Will put on a handpiece and put the biteblock
▣ The Usage Of The Product
This product is a mouth opening and lighting apparatus instead of dental light for the dentist for dental treatment and for her or his accurate clinical consultation of the patients(clients), and for the external work like home visit and emergency operation outside of clinic.
treatment and for her or his accurate clinical consultation of the patients(clients), and for the external work like home visit and emergency operation outside of clinic.
- Dental Light of dental treatment in dental clinic
- Dental consultation by the patient’s visit to the clinic:
medical examination of teeth and consultation of patient.
- Dental examination before and after implant, scaling and tooth whitening and reformation
- Dental examination by travel or by home visit, and external dental treatment for
the old or disabled person
▣ How To Autoclave The Product
• To use the Portable MaxBite Unit to the patient following the User’s Manual
• To separate BITEBLOCK from HANDPIECE.
To wash the BITEBLOCK in the flowing water.
Sterilize BITEBLOCK in the Autoclave.
* Autoclave sterilization regulation is subject to general stipulation on sterilization
ex) Sterilization at 135 C for 3minute, Sterilization at 120 C for 20 minute
* Do not wash or sterilize HANDPIECE, Lighting source in the Autoclave.
▣ How To Autoclave The Product
• Executive Dental does not admit the claim of the physical and chemical damages in the warranty caused by to your misuse beyond the usage of the product.
• Same way, the warranty does not cover the damage from the careless misuse of the product by your patients.
▣ How To Use Portable MaxBite Unit
• First, please check the size. There are large and small size for BITEBLOCK.
• Then, please check the using part of the product carefully if there is any damage on it.
If you find any damage, please stop to use it
• please insert illumination part to the end through the hole of BITEBLOCK
• Please insert BITEBLOCK of Portable eBITe Unit into the opposite side of the teeth
for the medical treatment
• Please push the power switch to operate Portable MaxBite Unit.
• Please do ultrasonic wash of the used BITEBLOCK.
• Finally please seal the used BITEBLOCK into the disinfected envelope
and sterilize it in the autoclave.
▣ How To Charge Battery
• To insert the plug of power cord into the socket
• To plug in the charging jack of adaptor into the HANDPIECE or CRADLE
• Battery charging is completed with the change of charge indicating lamp.
Charging time is around one(1) hours
▣ Composition Of Product
Please carefully check the structure of the product one by one before your installation.
• HANDPIECE with battery pack : 1ea
• BITEBLOCK : 1ea
• CRADLE : 1ea
• ADAPTOR with USB cable : 1ea
• User’s manual incorporated with warranty card : 1ea
▣ Attention To Your Use Of The Product
• Please separate HANDPIECE with battery pack for the safety and sanitation of the product.
• Every time after use of the product, please wash the BITEBLOCK and sterilize it.
• Please keep it in the sealed disinfected envelope.